3132_0 Micro Load Cell (0-780g) CZL616C
This load cell measures shear force up to 780g and connects to a bridge input.
Load cells are designed to measure force in one direction. They will often measure force in other directions, but the sensor sensitivity will be different, since parts of the load cell operating under compression are now in tension, and vice versa.
This Single Point Load Cell is used in small jewelry scales and kitchen scales. It’s mounted by bolting down the end of the load cell where the wires are attached, and applying force on the other end in the direction of the arrow. Where the force is applied is not critical, as this load cell measures a shearing effect on the beam, not the bending of the beam. If you mount a small platform on the load cell, as would be done in a small scale, this load cell provides accurate readings regardless of the position of the load on the platform.
Load cells require a power source and produce a very small voltage differential when under strain. In order to read this signal, an amplifier or a board with a high-precision analog-to-digital converter is required.
|Make sure to calibrate your load cell before using it. You can find information on how to calibrate the cell in the Load Cell Primer. You should also have a look at the User Guide for the bridge interface board that you're using to read the load cells.|
|Temp. effect on zero||0.05%F.S/10°C|
|Temp. effect on span||0.05%F.S/10°C|
|Compensated Temp. Range||-10°C~+40°C|
|Operating Temp. Range||-20°C~+55°C|
The Data Sheet below has a comprehensive glossary that describes in practical terms the meaning and usefulness of the specifications.