3513_0 CE-IZ02-32MS2-0.5 DC Current Sensor 0-1A
This DC Current sensor measures DC current, up to 1A. Powered by an external 12VDC power supply, the DC Current Sensor provides a 0 to 5VDC analog signal that is linearly proportional to the current measured at the terminal blocks.
The DC Current Sensor can easily be mounted using the screw holes, or on a 35mm wide DIN rail for quick installation. The terminal blocks accommodate conductors of up to AWG #16.
In order to measure the current, the sensor has an internal 0.5 Ohm resistor. This will result in a slight voltage drop before the voltage gets to the device requiring the power. Consequently, the “leftover” voltage might not be high enough to power the device, especially if the power supply has a voltage close to the device’s minimum voltage requirements. As a point of comparison, using a 12V battery running a 16 Ohm Load, there is an approximate drop of 0.375V due to the 0.5 Ohm sensing resistor on the 100mA DC Current Sensor. The maximum voltage drop is 0.5V, when the sensor is measuring 1000mA.
The output of this sensor is not clamped at 5V. If the current being measured exceeds the rated input current, more than 5V will appear at the output signal. This can damage the Phidget Board (or any other ADC) that it is attached to if the output signal exceeds 5V for an extended period of time.
The current sensor connects to an analog input on Phidgets I/O boards.
The formula to translate the SensorValue into Current is: I(mA) = SensorValue
- This sensor measures up to 1A DC and connects to an analog input
For boards that have power input, such as the 1019 or the 1072, if the supplied power is 12V, then the + terminal block on the Phidget Board can be connected to the voltage sensor terminal 5 (+12V), and the G terminal block on the Phidget Board can be connected to the sensor terminal 6 (-).
This sensor connects to any device with an Analog Input.
You can use the 3002 - Sensor Cable 60cm or the 3004 - Sensor Cable 350cm to connect the DC Voltage Sensor to your Phidget. Just snip off the connector from one end of the sensor cable and connect the white wire to the terminal marked with an 8, and the black wire to the terminal marked with a 6. The red wire is not used. You can also use a 3031 – Female Pigtail.
This is how you connect the 3512 to a Phidget Board that does not have 12V power supply, such as the 1018 or 1203, using a Sensor Cable.
- Connect the Power Supply +12V wire (the wire with the white line) to the Current Sensor terminal 5 (+12V).
- Connect the Power supply Ground wire and the Analog Sensor Cable Black wire to the Current Sensor terminal 6 (-).
- Connect the sensor cable white wire to the Current Sensor terminal 8 (+).
- Connect the positive wire of the device’s power supply to the Current Sensor terminal marked 1 (IN+).
- Connect a wire from the input voltage of the device under test to the Current Sensor terminal 3 (IN-).
- For boards that have power input, such as the 1019 or the 1072, if the supplied power is 12V, then the + terminal block on the Phidget Board can be connected to the voltage sensor terminal 5 (+12V), and the G terminal block on the Phidget Board can be connected to the sensor terminal 6 (-).
|Sensor Type||Current (DC In-Line)|
|Sensor Output Type||Non-Ratiometric|
|Input Current Min||0 A|
|Input Current Max||1 A|
|Measurement Error Max||0.5 %|
|Sensor Response Time Max||15 ms|
|Supply Voltage Min||12 V DC|
|Supply Voltage Max||12 V DC|
|Current Consumption Max||16.7 mA|
|Output Voltage Min||0 V DC|
|Output Voltage Max||5 V DC|
|Isolation Voltage (DC)||2.5 kV DC|
|Isolation Voltage (AC)||1.5 kV AC|
|Recommended Wire Size||12 - 24 AWG|
|Operating Temperature Min||0 °C|
|Operating Temperature Max||50 °C|