The DC Current Sensor can easily be mounted using the screw holes, or on a 35mm wide DIN rail for quick installation. The terminal blocks accommodate conductors of up to AWG #16.
In order to measure the current, the sensor has an internal 50 Ohm resistor. This will result in a slight voltage drop before the voltage gets to the device requiring the power. Consequently, the “leftover” voltage might not be high enough to power the device, especially if the power supply has a voltage close to the device’s minimum voltage requirements. As a point of comparison, using a 9V battery running through an LED with a forward voltage of 1.8V and a 1000 Ohm resistor, there is an approximate drop of 0.35V due to the 50 Ohm sensing resistor on the 10mA DC Current Sensor. The maximum voltage drop is 0.5V, when the sensor is measuring 10mA.
The output of this sensor is not clamped at 5V. If the current being measured exceeds the rated input current, more than 5V will appear at the output signal. This can damage the Phidget Board (or any other ADC) that it is attached to if the output signal exceeds 5V for an extended period of time.
The current sensor connects to an analog input on Phidgets I/O boards.
The formula to translate the SensorValue into Current is:
I(mA) = SensorValue * 0.01
- Measures DC Current up to 10mA
- 0.5% Accuracy
- Isolation Voltage of 2500VDC
- DIN Rail mountable
- Connects to any device with an Analog Input:
Devices with Analog Inputs
|Product #||Name||# of Analog Inputs||Additional Features|
||PhidgetInterfaceKit 8/8/8 Mini-Format
||DIP-36 Package for compact and OEM applications, 8 Digital Inputs, 8 Digital Outputs
||Compact USB Dongle Size, 2 Digital Inputs, 2 Digital Outputs
||8 Digital Inputs, 8 Digital Outputs
||PhidgetInterfaceKit 8/8/8 w/6 Port Hub
||8 Digital Inputs, 8 Digital Outputs, 6-port USB Hub
||1 DC Motor control, 2 Digital Inputs, 1 Encoder Input
||Single Board Computer, 8 Digital Inputs, 8 Digital Output, 6-port USB Hub
||LCD Character Screen, 8 Digital Inputs, 8 Digital Outputs
For boards that have power input, such as the 1019 or the 1072, if the supplied power is 12V, then the + terminal block on the Phidget Board can be connected to the voltage sensor terminal 5 (+12V), and the G terminal block on the Phidget Board can be connected to the sensor terminal 6 (-).
||Current (DC In-Line)
|Sensor Output Type
|Input Current Min
|Input Current Max
|Measurement Error Max
|Sensor Response Time Max
|Supply Voltage Min
||12 V DC
|Supply Voltage Max
||12 V DC
|Current Consumption Max
|Output Voltage Min
||0 V DC
|Output Voltage Max
||5 V DC
|Isolation Voltage (DC)
||2.5 kV DC
|Isolation Voltage (AC)
||1.5 kV AC
|Terminal Block Size
|Operating Temperature Min
|Operating Temperature Max
This sensor connects to any device with an Analog Input.
You can use the 3002 - Sensor Cable 60cm or the 3004 - Sensor Cable 350cm to connect the DC Voltage Sensor to your Phidget. Just snip off the connector from one end of the sensor cable and connect the white wire to the terminal marked with an 8, and the black wire to the terminal marked with a 6. The red wire is not used. You can also use a 3031 – Female Pigtail.
This is how you connect the 3511 to a Phidget Board that does not have 12V power supply, such as the 1018 or 1203, using a Sensor Cable.
- Connect the Power Supply +12V wire (the wire with the white line) to the Current Sensor terminal 5 (+12V).
- Connect the Power supply Ground wire and the Analog Sensor Cable Black wire to the Current Sensor terminal 6 (-).
- Connect the sensor cable white wire to the Current Sensor terminal 8 (+).
- Connect the positive wire of the device’s power supply to the Current Sensor terminal marked 1 (IN+).
- Connect a wire from the input voltage of the device under test to the Current Sensor terminal 3 (IN-).
- For boards that have power input, such as the 1019 or the 1072, if the supplied power is 12V, then the + terminal block on the Phidget Board can be connected to the voltage sensor terminal 5 (+12V), and the G terminal block on the Phidget Board can be connected to the sensor terminal 6 (-).
Recommended Power Supplies: